There are control points  on the abutment which can be moved inwards, outwards, up and down to change the abutment's shape. Note that the software ensures that undercuts are avoided, and production-specific design restrictions are enforced. Therefore, there is - intentionally - some restriction in the movement, and moving one control point can cause another control point above or below to move as well.
With the green control point  between the arrows in the center of the abutment, you can adjust the overall abutment height.
Wizard tab: Top
The slider  in the wizard window enforces a minimum draft angle of the abutment, and it doesn't allow the abutment to have undercuts. Lower values are easier to design.
It is also possible to tell the software to enforce a certain space between the anatomy and the abutment with the Spacing slider . If Auto Adapt checkbox  is set, both the distance and the occlusal shape of the abutment are precisely offset from the chewing surface.
If a change to  or  is entered, these changes will only be applied if either the Apply  or the Reduce only  button is clicked. The Apply  button will both reduce and grow the shape in order to reach the desired distance values; the button Reduce only  will only shrink the abutment at parts where the distance to the anatomy is lower than the set value.
With activating the checkbox Distance to anatomy , you can visualize the distance of the anatomy to the abutment, the visualization is displayed on the abutment. The visualization is color-coded from red (low distance), to blue (high distance).
The highest value is set by the slider Spacing .
You can also have a measurement Distance to abutment  displayed on the anatomy.
Wizard tab: Bottom
Use this tab to correct your abutment bottom design, without going backward in the wizard. For details, see section "Generate abutment bottoms"
Wizard tab: Advanced
Use this tab to adjust the profile borders, angulate screw channels, set the milling parameters, and adjust the margins.
The slider Height  sets the minimum height of the abutment profile border.
The slider Radius  controls the round-off at the sharp edge of the border.
Allow below implant  gives you the ability to drag border control points below the surface of the implant.
CAUTION: USE ONLY IF YOU KNOW WHAT YOU ARE DOING!
: Allows you to move all points that define the border (abutment margin) at the same time, inwards, outwards, up or down.
: These controls allow you to stick or unstick all of the border control points to the gingiva (tissue).
Angulated screw channels
Angulated screw channels  is possible, if the configuration of your exocad software, and the implant library used, allows this.
There are two possibilities to do it:
Click anatomic surface, pre-op or waxup scan to let the screw channel go through this point.
Hint: Hold CTRL and click to reset the previously manually defined angle.
Define screw channel angle by dragging an arrow. Holding CTRL and SHIFT restricts angle movement to the current angulation plane.
Holding CTRL and 'X' restricts angle movement to the implant analog plane.
Holding CTRL and 'I' snaps the angle to the nearest Integer value.
Holding CTRL, SHIFT and 'I' snaps the angle to intervals of 1°.
See our video tutorial for details.
Expanding the Milling Parameters  control reveals the following:
-  Diam. can be used to set the diameter of the tool used to mill the abutment, also known as the tool compensation. You should add at least 0.2mm to the smallest tool size.
-  Tool sets the diameter of the tool used to mill the suprastructure (crown or coping).
-  Min thickness sets the minimum thickness of the abutment's axial walls.
-  Screw Distance keeps a minimum distance of the abutment to the screw channel. The control points can't be set closer to the screw channel than this value.
-  Margin is the minimum angle of the margin of the abutment.