Designing a smile with the Smile Creator Wizard
To start Smile Creator, click Start Smile Creator in the Smile Creator Wizard. The wizard shows the step “Start Smile Creator” if at least two neighboring teeth are reconstructed and at least one of them is an incisor.
Smile Creator automatically detects facial features with artificial intelligence to automate the smile design and get to an esthetic proposal faster. This, along with many other new features and improvements, helps saving valuable time while designing cases.
Note: Smile Creator is not available if the reconstruction type is a waxup with a waxup scan.
Alternatively, start Smile Creator from the main toolbar (Tools > Smile Creator)
Note: The Smile Creator module first needs to be activated. Also note that Smile Creator is optimized for single groups of neighboring teeth.
Load a retracted image
Click Load a retracted image to load an image from your file directory, or take a picture directly from your device:
1. Click the camera symbol.
2. Click Take image in the window that opens.
Alternatively, you can drag&drop an image file into the main view, or copy&paste an image file into the main view.
Set corresponding points and adjust
DentalCAD automatically aligns the image to the scan data. If this process fails, the patient’s photo is displayed on the right side of the screen. Match the scan and the photo by setting alignment points.
Set two corresponding alignment points on the left and right image. First click two points in the scan and then click the corresponding alignment points in the corresponding positions in the photo.
Adjust the point positions as required by left-clicking and dragging them. Hover the mouse over the images and scroll to magnify them for better adjustment. Click Next when the points are optimal.
In the next step, the scanned jaw and retracted image are placed on top of each other.
Click Next when done.
Load a smiling image
Next, load a smiling image of the patient by clicking Load a smile image and selecting an image from your file directory, or take a picture directly with your device. Alternatively, you can drag&drop an image file into the main view, or copy&paste an image file into the main view.
Set corresponding points and adjust
Load a face scan
Draw a lip line
DentalCAD automatically detects the lip line with artificial intelligence. If this fails, draw a lip line to cut out the mouth by clicking points along the inner lip line, as shown in the screenshot below. Adjust the line as required by dragging the points.
Guidelines and curves
Also using artificial intelligence, DentalCAD automatically detects the patient’s eyes. If this fails, follow these steps:
1. Click the patient’s right eye.
2. Click the patient’s left eye. Guidelines and curves appear. Helpers
3. In the wizard, select a proportion guide by clicking the drop-down menu  .
4. To add more guidelines, expand the Add lines and curves section  in the wizard and click the corresponding symbol.
5. Adjust guidelines and curves as required by dragging and dropping their control points. Click a control point and press DEL to delete the guideline.
6. Align the image by enabling Lock Rotation and align image to lines .
7. Click Next when done.
Next, adjust your design. The main window shows the retracted image with the 2D outlines of the tooth models. In the upper right corner is the front view of the 3D model. The top view is shown below and the image in the lower right corner represents the side view of the model. You can activate the Show Distances function in the main toolbar.
The Smile Creator Wizard is shown in the left part of the screen. The four areas of the window will be explained in this section.
Select a dental library in the Library section .
Note that it is important to begin the modeling process with this step. Changing the dental library resets any modifications you have made.
Check the Mirror Movements  box to apply all changes in position, scaling and rotation symmetrically. In the Tools section  of the wizard you can click a tooth to view and change its dimensions in all views. You can select several teeth by holding CTRL while clicking an additional tooth. All teeth in between two clicked teeth will also be selected when holding SHIFT while clicking. Select either Free-Form or Measurements to perform advanced modelling. Click Free-Form to edit the tooth shape. Check this page for further information about this tool.
Select Measurements  to view the dimensions of a tooth and the proportions. Disable the function by clicking the icon again.
Check the Lock view  orientation box to lock the current view. Check the Auto hide view obstructions  box to automatically hide parts of the scan obstructing the view in the three secondary views. In the Visual properties , select one of the following modes:
- Full tooth: displays all teeth in their entirety with the tooth necks in realistic rendering
- Preview: displays a preview of the teeth above the cervical border (useful for so-called “gum smiles”) in realistic rendering
- Outline: displays an outline of the teeth above the cervical border.
Click Undo to cancel previous changes and Redo to restore them.
Note that changes to the tooth library will also be undone/redone.
Adjust the model directly in the viewer, if required, as follows:
Click a tooth in any of the views to modify its size, shape or position.
- Drag the frame that appears by its control points to change the size of a tooth.
- Drag the control point with arrows to rotate the tooth.
- Click a tooth to change its position freely. To perform this step for several teeth simultaneously, hold CTRL and click the desired teeth or click, hold and drag the mouse to create a rectangular selection area for several teeth.
- Switch between the outline view and full tooth view by clicking the circle icon in the upper right corner of each view.
- Rotate the model freely by right-clicking and dragging it. Click the arrow in the top right corner return to the default view.
- Click Next when done.
Preview and Color
In this step, a preview of the smile is displayed. There is a before/after slider to compare the previous and the new smile with a slider dragging the arrows to the left or right.
You can also adjust the coloring of the teeth using the following controls:
- Click Use 2D color bar  to open a custom color window that allows you to choose the optimum tooth color from the palette. Choose the Tooth base color and Tooth highlight color with the color picker.
- Select one of the 20 standard colors in the Wizard. Fine-tune your selected color with Adjust tooth color  to better match the photo using the slider below.
Note that the color in the photo depends on the camera, lighting and display configuration, so the preview is only an approximation.
- Expand Custom Color Selection to give your design an individual coloration. Define the tooth base color and the tooth highlight color by using the sliders.
Use the color picker to select a color in the photo by clicking the upper pipette icon. Click a point in the photo which corresponds to the tooth base color.
Next, click the lower pipette icon to select a highlight color. Click a bright position in the original image to apply this as the highlight color for your model.
Fine-tune the Image brightness with the corresponding slider.
- Select either Full Tooth, Preview or Outline mode using the pick list towards the bottom of the wizard.
-  Export the New smile design only or the new and previous smile coping to the clipboard, or save directly in the project directory.
Smile Creator Report
In the last step, you can create a PDF report of the smile design. In the Page setup configuration , choose the Size, type of margin, Orientation, and Theme according to your preference. Below you will find an example:
Select individually what you would like to include within the Print all content expander .
- Click Done  when you are finished.
- After finishing the Smile Creator, it is available the Smile window, a secondary view to show the patient-oriented smile design. This secondary view can always be opened again during subsequent DentalCAD workflow and it's available on the Main Toolbar. This will allow users to use advanced tools like the chain-mode tooth placement as usual, but at the same time see the outcome interactively relative to the patient's face. Use this tool within the same screen or with a multiple screens setup.